The Bagley correction is needed to calculate the inlet pressure drop when round hole capillaries are used. Near the inlet area of round hole capillaries, a pressure drop is resulting from convergent flow in the capillary inlet. In the same way a part of the pressure is also used for the outlet. The Bagley correction is supporting the separation of the viscose pressure drop and the inlet/outlet pressure drop. To determine the inlet/outlet pressure drop the pressure drop is being applied with different capillaries of the same diameter but different lengths and extrapolated to zero (Bagley plot).
For highly viscous materials (> 1 Pas), e.g. plastics and rubbers, the Bagley correction is the correction that will be applied first. After the Bagley Correction has been carried out the Rabinowitsch-Weissenberg correction can be carried out. This sequence must be adhered for physical reasons. The GÖTTFERT software WinRheo II is covering the order of the corrections in an automatic way.
Send us an e-mail firstname.lastname@example.org or call us +49 (0) 62 81 408 - 0
We will help you with pleasure.